|Date Added:||4 May 2016|
|File Size:||36.65 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I am looking for the same solution. Click the Tektronix project to create a new copy. Create a constant for the new extension terminal on the left side of the Replace File Extension Node. The Peak Amplitude Node reads in the waveform signal that the Read Multiple Waveforms Node creates and analyzes the data your Tektronix device collects. Calculate yourself the number of bytes transerred for both cases. This tool automatically detects USB hardware connected to your host PC and adds a graphical representation to the window.
Create a constant for the type terminal on the top left side of the Get System Directory Node.
Tektronix DPO Digital Phosphor Oscilloscope – Instrument Driver – National Instruments
Message 3 of 5. Right click the Vertical Range constant and choose Change to control.
Furthermore it takes a whole lot of time to transfer the picture. Open the Learning tab by selecting the Learn to Program tile in the Welcome screen or selecting the Learning tab from the top right.
Tektronix MDO MSO DPO Series – National Instruments
The instrument is now configured. Before you run the gvi type in the desired file name, save the lsbview in the default LabVIEW directory and you can see the new file path in the New File Path indicator. These instruments also feature numerous triggering modes, deep onboard memory, and a driver software API that includes data streaming and analysis functions.
The NI Instrument Driver Network houses thousands of instrument drivers for automating third party instruments. You can easily sort the Diagram by clicking and dragging to help with reading and understanding program flow. You can then place items one by ,abview on the Panel. Ethernet network devices have to be added using the IP address the device is given.
With the File Name control, you can name the new. Message 2 of 5. Rename the control “File Name”.
Then click the green Run arrow at the top left of the VI. With the File Name control, you can name the.
Customize the Panel of the Dpoo4054 to fit your applications needs. Delete the yellow Error wire between the two. The instrument is then detected.
Delete the Yellow Error Wire. Keeping controls on the left of the Panel and indicators on the right helps user readability.