The case a you don’t have to change the PC baudrate manually and the instrument baudrate must only be the same with it. Setting comm baudrate may have two cases — 1 set baudrate at PC side, 2 set baudrate at instrument side. GPIB has no such baudrate values. Also a clear and simple way to do relative measurements to subtract cable and attenuator losses. There is no default logical name for EA unless you create one. This application brief demonstrates the capabilities Keysight power meters and power sensors have for measuring CW and average power signals.
|Date Added:||24 June 2005|
|File Size:||15.21 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Most connected devices provide a way to program scripted test automation.
Once you created a logical name, you can pass it to the IVI-COM Session Factory to indirectly create a driver instance, or you can pass it to directly to the specific driver’s Initialize method. Most Active Hardware Boards: But normally no meaning to use or slower baudrate with your Pentium4 PC, if the instrument is supporting faster baudrates.
Message 8 of Message 3 of After buying measurement device meteer multiple thousands of euros, you also have to pay monthly subscription fee to use your legally owned measurement device.
Keysight UA in measurement. Also you have to pay subscription fee. Select a Web Agilwnt Choose a web site to get translated content where available and see local events and offers.
Measuring Agilent Power Meter using VC++ – Discussion Forums – National Instruments
It also includes comprehensive selection charts and a compatibility table to help you pick out the right power meter and sensor. The case b you can use any baudrate you want as long as the NI-MAX configuration and instrument baudate are matched.
Trial Software Product Updates. This application note describes some useful tips on how to effectively minimize test times while obtaining power measurements.
I also added some features that I felt, were missing in the gigantic BenchVue application. Based on your location, we recommend that you select: As all USB connected measurement devices, this power meter needs a software to display the measurements.
Measure Power on a Keysight RF Power Meter using the IVI-C Driver – MATLAB & Simulink Example
See search results instead: Can somebody help me in this??? Previous 1 2 3 4 Next. You don’t have to ibi this command when using GPIB.
Yes it is the right way. But if the app wants to change it through the online, you will use SYSTem: I have read about using IVI session factory Message 7 of It was an expensive, but is a nice device. Agilwnt 4 of U Series Firmware Useries Firmware file and documentation.
Also, I wanted to know if you could communicate to the instrument at all, try doing just a simple query to see if you can. MathWorks does not warrant, and disclaims all liability for, the accuracy, suitability, or fitness for purpose of the translation.
If the app is unable to communicate with the instrument due to unknown baudrate, it is also imposiible to send SYSTem: GPIB aagilent no such baudrate values.
The automated translation of this page is provided by a general purpose third party translator tool. Posted on agilemt, This page has been translated by MathWorks. Choose a web site to get translated content where available and see local events and offers.
As many electronics software – it is Windows only.